A. Multi-Product Calibrator. Service Manual. PN August Rev.5 , The Fluke Model A Multi-Product Calibrator (Figure ) is a precise. Fluke Corporation. 1. A Specifications. The following paragraphs detail specifications for the A Calibrator. changes more than 5 °C. (See “Zeroing the Calibrator” in Chapter 4 of the A Operator Manual.) Also see additional. 2 Fluke Corporation A/A Multi-Product Calibrators Extended Specifications. Warmup Time. Twice the time since last warmed up, to a maximum of
|Published (Last):||12 April 2018|
|PDF File Size:||20.14 Mb|
|ePub File Size:||14.22 Mb|
|Price:||Free* [*Free Regsitration Required]|
Press the [reset] key. Eor each measurement, take samples for at least two seconds, using the HP 8 A MATH functions to determine the average flukw mean value. Edge Specifications Table You are then prompted to enter into the A the value read on the measurement instrument.
Adjusting the Wave Peak Center with R omf. Complete the following procedure to zero the calibrator. If the warning still occurs, repair may be necessary. The location of fluke 5500a calibrator manual internal fuses are summarized in Table Thermocouple Fluke 5500a calibrator manual Accuracy The Thermocouple Sourcing Accuracy test checks the accuracy of the thermocouple measuring circuitry.
Enter the LCR reading into the A when prompted. This test exposes failures at full scale positive.
If your SC is not installed or not recei ving p ower, the following error message appears on the display when you press scoPE to access the oscilloscope calibration menus. Enter the actual fluke 5500a calibrator manual Q. FR A7 Overcurrent fault Overload functionality Procedure provided in this manual.
Equipment Required for Caiibration and Verification Fluke 5500a calibrator manual equipment listed in Table is required to calibrate and verify performance of the A. Ensure the HP A reading is 0. AC Voltage Sinewave Specifications Connections for Caiibrating TC Measure 1. Allow the A reading to stabilize, then enter the reading amnual Column A of the table.
Resistance Accuracy Test cent. It is possible to achieve a very fluke 5500a calibrator manual ledge close to the horizontal center, but if the peak is too high or too low, then the aberrations will not be properly adjusted.
Use this same range for the corresponding baseline measurements at each step. Adjusting the Leveied Sine Wave Fluke 5500a calibrator manual Note This procedure should only be used for adjusting the leveled sine wave harmonics. Repeat steps 4 and 5 for all of frequencies listed in Table Access Procedures Use the following procedures to remove the following assemblies: For harmonics 3, 4, and 5, record the highest harmonic level of the three measured.
DACS are used to control the level of dc fluke 5500a calibrator manual and to control the amplitude of ac signals. Repeat steps 5 to 7 until the Calibrator Mainframe display prompts to connect a resistor. High frequency at 5. Refer to Figure Adjusting the Aherrations for the Edge Function Page of Go.
Input sense high to AUX high.
Fluke A Calibrator – No Options | eBay
Returns the calibrator to the fluke 5500a calibrator manual state: Adjusting the Aberrations for the Edge Function General specifications that apply to the Calibrator Mainframe can he found in Chapter 1. By pressing ‘print’ button you will print only current page.
Readjust A90R36 and A90R35 to obtain equal amplitudes for the first, second, and third aberrations. All the transformer connections for the outguard supplies come through one bundle of wires connected to the motherboard with PI. Then follow these steps to calibrate Leveled Sine Fluke 5500a calibrator manual amplitude.
A Service Manual
One of these cables is fastened by fluke 5500a calibrator manual cable tie that must be cut, then replaced with a new one when reassembling.
Specifications The following paragraphs detail specifications for the A Calibrator. Tunnel Diode Pulser Amplitude Verification The following equipment is used to verify the edge rise time.